View recent webinars top top "Techniques for enhancing Visualization of Lyophilization Cakes"; the three-part series on "Lyophilization Validation: operational Qualification and procedure Validation"; discover the "Basic theory of frozen Drying" and much more...
You are watching: How much time do manufacturers suggest for proper system evacuation?
SP company professionals offer your organization a depth of field of expertise that is truly unique to ours selected industries and markets. We room recognized experts in applications, methodologies, rap processes and many other areas that directly influence your day-to-day operations. V SP your systems is simply a speak to or one e-mail away.
SP company professionals are extremely knowledgeable and are default to assistance every instrument, software and application product the SP has brought to the market.
SP tools you have come to know and trust - SP FTS, SP Genevac, SP Hotpack, SP Hull,SP PennTech, SP Virtis and also SP i-Dositecno.
It"s *Your Time, her Investment, and also Your Results* - allow our technological sales applications specialists help you select the exactly lyophilization and scientific tools configuration for her application. SP specializes in frozen drying/lyophilizataion, complete aseptic fill-finish production lines, centrifugal evaporation and also concentration, temperature control/thermal management, glassware washers and also controlled environments.
HOME>Learning Center>Tech Briefs>Refrigeration mechanism Evacuation - 2012-09
Refrigeration device Evacuation - 2012-09
Refrigeration system Evacuation
Whenevera refrigeration mechanism is opened up for service, appropriate evacuation is necessarybefore it can be safe resealed. That is an essential that non-condensables andother contaminates be removed from the system.
Asuitable vacuum pump, one qualified of blank-off come at the very least 300 microns orlower, should be linked to both the high and also low political parties of the refrigerationsystem. The dimension of the connecting hoses should be such the they will notrestrict the flow from the system to the vacuum pump. A vacuum gauge that readsin microns need to be connected to the furthest point in the mechanism away native vacuum pump.
Atriple evacuation procedure is strongly recommended. Because that triple evacuation, pumpdown the refrigeration system to 1,500 microns, and then break the vacuum usingdry nitrogen. At 1,500 microns any kind of moisture or ice trapped in the mechanism willoutgas. ~ backfilling through dry nitrogen to atmospheric pressure, operatethe vacuum pump a 2nd time to 1,500 microns and again backfill with drynitrogen. Finally, operate the vacuum pump the third time come 300 microns, butno lower. Close every valves and isolate the vacuum pump, then rotate the vacuumpump off.
Caution: at a pressure below300 microns (mT) the POE oil in the compressors will start to degrade and beginlosing that is lubricating ability.
Watchthe vacuum gauge to ensure vacuum is holding. If after five minutes there is aslight ns of vacuum, there might possibly be some residual outgassing in thesystem. Listed below 1,500 microns any remaining moisture is present as ice which willsublime. In this case the vacuum pump need to be operated one an ext time tofurther dry the system. After a organize time the ten come fifteen minutes at 300microns the mechanism is taken into consideration successfully evacuated.
Aninability to pump under to 1,500 microns indicates a device leak or a pumpproblem. A loss of vacuum to above 1,500 microns throughout the host test indicatesa mechanism leak. Device leaks should be repaired prior to the refrigeration systemcan be safely operated. Any system leak needs you go through the necessarysteps come insure there has been no contamination of the refrigerant.
Somethingto remember: ~ you have actually finished making use of the vacuum pump, a great procedureis to readjust the oil. Any type of contamination in the refrigeration system is now inthe vacuum pump oil. If you perform not change the oil and also the vacuum pump sits idlefor any period of time, the air pollution will start attacking its inner components.
See more: Which Two Hemispheres Is The United States Located, The 4 Hemispheres Of The World
Learn more about her Freeze Dryer - to visit our TrainingSeminarsTech advice Home
These technological tips are for usage by qualified maintenancepersonnel that are familiar with their details freeze dry equipment. Thesetips room intended just as general guidelines. Lyophilization equipment isfrequently custom configured and some tips might not be suitable for allfreeze dryers. Always read and also follow the directions of your equipment"smaintenance manual. If friend would choose to talk about one that our technology tips, pleasecontact Dave Clayton in ~ 215-672-7800 ext-1376.